This article is only available in the PDF format. Download the PDF to view the article, as well as its associated figures and tables.
Ever since Dr. W. A. Lane introduced the use of metal plates in the open reduction and fixation of fractures there has been some kind of bone clamp used to hold plate and fractured parts in place. This has been done with more or less difficulty and inconvenience to the operator.
I have devised an instrument which I believe is an improvement and which has proved very satisfactory to me. It consists of two long handles with lock effect about 2 inches from the clamp proper. In the upper handle at the distal end it has a bar attached, which has two prongs at the lower portion. The upper portion extends about 2½ to 3 inches upright for the purpose of carrying the shuttle. The shuttle carries the upper prong at its lower portion to complete the clamp. The shuttle may have two prongs and is interchangeable. In the
Rush HP. NEW BONE-HOLDING CLAMP. JAMA. 1937;108(25):2118. doi:10.1001/jama.1937.92780250001007a
Coronavirus Resource Center
Customize your JAMA Network experience by selecting one or more topics from the list below.
Create a personal account or sign in to: