This article is only available in the PDF format. Download the PDF to view the article, as well as its associated figures and tables.
To the Editor:—
In an artificial kidney not all of the area of the cellophane tubing takes part in the dialysis. It is important to know what the effective dialysis area is of a specific artificial kidney.Evans blue dye added to the dialyzing fluid of an artificial kidney stains the cellophane on contact (5 ml of 1% Evans blue per 100 liters). Removal and drying of the cellophane from the dialyzer provides a permanent colorimetric record that indicates where supporting structures blocked the cellophane. The faults in the support, whether ridges, foam, or screens, are demonstrated ( perhaps they are correctable), and the percentage of area available for dialysis can be calculated (Fig 1).When Evans blue has been added to the bloodstream of an artificial kidney, the cellophane not colored in a certain area reveals permanently the places through which no blood would flow. Total compression of the cellophane
Erben J, Kvasnicka J, Groh J, Rose F, Kolff WJ. Colorimetric Determination Of Effective Dialysis Area Of Artificial Kidney. JAMA. 1967;202(1):72. doi:10.1001/jama.1967.03130140130034
Customize your JAMA Network experience by selecting one or more topics from the list below.
Create a personal account or sign in to: