This article is only available in the PDF format. Download the PDF to view the article, as well as its associated figures and tables.
To the Editor.—
The Food and Drug Administration recently investigated an incident involving an infant death attributed to a malfunctioning neonatal incubator. An Air Shields C86 Isolette incubator overheated owing to two unrelated failures: (1) The primary thermostat functioned intermittently, ie, most of the time the thermostat would function properly, turning the heater off and on to maintain the incubator temperature at the set-point temperature; but occasionally, the thermostat would "stick" in the "on" position, allowing the heater to function uncontrolled. (2) A back-up safety thermostat (mercury in glass type) was broken. Under normal conditions when the temperature reached 39.4 °C, this thermostat would have turned off the heater and sounded an alarm.The FDA investigation also revealed that the primary thermostat could exceed its recommended maximum temperature (37.2 °C) if allowed to drift out of calibration. Analysis shows further that backup safety thermostats can malfunction from mercury column separation
Haffner ME. Malfunction of a Neonatal Incubator. JAMA. 1982;247(17):2372. doi:10.1001/jama.1982.03320420028023
Customize your JAMA Network experience by selecting one or more topics from the list below.
Create a personal account or sign in to: