Customize your JAMA Network experience by selecting one or more topics from the list below.
Reed NS, Betz J, Kendig N, Korczak M, Lin FR. Personal Sound Amplification Products vs a Conventional Hearing Aid for Speech Understanding in Noise. JAMA. 2017;318(1):89–90. doi:10.1001/jama.2017.6905
Presently, hearing aids can only be purchased in the United States through a licensed professional, with a mean cost of $4700 for 2 hearing aids (uncovered by Medicare).1-3 According to nationally representative estimates based on 2605 adults from 1999 through 2006, less than 20% of adults with hearing loss report hearing aid use.4 Personal sound amplification products (PSAPs) are less-expensive, over-the-counter devices not specifically labeled for hearing loss treatment, but some are technologically comparable with hearing aids and may be appropriate for mild to moderate hearing loss.1 We compared a sample of these devices with a conventional hearing aid among individuals with mild to moderate hearing loss.
This study protocol was approved by the institutional review board at Towson University and Johns Hopkins, and written informed consent was obtained from each participant. From April 2016 through January 2017, we screened adults aged 60 to 85 years at a university audiology clinic to recruit a convenience sample for a randomized study. Inclusion criteria included mild to moderate hearing loss (20-55 dB HL; pure-tone average, 500-4000 Hz), no prior amplification use, and no cognitive impairment (Mini Mental State Examination score, >24).
Participants completed the AZBio sentence-in-noise task,5 a routinely used measure of speech understanding to assess functional hearing. Participants repeated sentences in the presence of background noise under 7 conditions: unaided, with a hearing aid, and with 5 PSAPs. To control for order effects, sentence lists and devices were randomly ordered using a Latin square design balancing first-order carryover effects. Accuracy (percentage of words repeated correctly; range, 0%-100%) was recorded for each condition (20 sentences per condition).
All testing was completed in a calibrated sound booth by an audiologist. To simulate a moderately difficult listening environment, sentences were presented via a speaker at a 0° azimuth (ie, directly in front of the participant) at 35 dB HL while speech babble noise was concurrently presented at a 180° azimuth (ie, directly behind) at 30 dB HL. A sample of 9 of the most-sold PSAPs via a large e-commerce retailer was assembled and tested for electroacoustic properties. Of the 9, the 4 PSAPs with the most favorable electroacoustic properties and 1 PSAP available in retail pharmacies were chosen and compared with 1 hearing aid commonly dispensed in a university audiology clinic. The same units were tested in each participant, unilaterally fit and adjusted to each participant’s hearing in their better-hearing ear using best-practice verification methods by an audiologist.
Linear mixed-effects regression models were used to model the within-participant change in performance with each device. Analyses were performed using R (R Foundation), version 3.3.2.
Of 63 adults screened, 42 met inclusion criteria (mean age, 71.6 years; women, 67%). The change in accuracy in speech understanding from unaided to aided varied by device (Table). The hearing aid and 4 of the PSAPs improved speech understanding from the unaided condition. The mean unaided accuracy was 76.5%. The hearing aid improved speech understanding accuracy to 88.4% for an absolute improvement difference of 11.9 percentage points (95% CI, 9.8 to 14.0). Three of the PSAPs demonstrated an improvement that was within 5 percentage points of the hearing aid (Sound World Solutions CS50+: accuracy, 87.4%, difference, 11.0 percentage points [95% CI, 8.8 to 13.1]; Soundhawk: accuracy, 86.7%, difference, 10.2 percentage points [95% CI, 8.0 to 12.3]; Etymotic BEAN: accuracy, 84.1%, difference, 7.7 percentage points [95% CI, 5.5 to 9.8]). For 1 PSAP, mean aided speech understanding was worse than the unaided condition (MSA 30X: accuracy, 65.3%, difference, −11.2 percentage points [95% CI, −15.2 to −7.3]).
Select PSAPs were associated with improvements in speech understanding for individuals with hearing loss that were similar to results obtained with a hearing aid, whereas 1 demonstrated little improvement and 1 degraded speech understanding. To our knowledge, this is the first study of PSAPs that controlled for between-participant variability and order effects. The study was limited by a modest number of participants, sampled by convenience, who were tested in a controlled audiological setting under unilaterally aided conditions and with only a limited sample of currently available hearing technologies. Whether similar results would have been obtained with other devices or if the user self-programmed the device is unknown. Results lend support to current national initiatives from the National Academies,1 White House,2 and bipartisan legislation3 requesting that the US Food and Drug Administration create a new regulatory classification for hearing devices meeting appropriate specifications to be available over the counter.
Accepted for Publication: May 10, 2017.
Corresponding Author: Nicholas S. Reed, AuD, Johns Hopkins Center on Aging and Health, Department of Otolaryngology—Head and Neck Surgery, Johns Hopkins School of Medicine, 2024 E Monument St, Ste 2-700, Baltimore, MD 21205 (firstname.lastname@example.org).
Author Contributions: Dr Reed had full access to all of the data in the study and takes responsibility for the integrity of the data and the accuracy of the data analysis.
Concept and design: All authors.
Acquisition, analysis, or interpretation of data: Reed, Betz, Kendig, Korczak.
Drafting of the manuscript: All authors.
Critical revision of the manuscript for important intellectual content: Reed, Betz, Lin.
Statistical analysis: Reed, Betz, Kendig, Lin.
Obtained funding: Lin.
Administrative, technical, or material support: Reed, Betz, Lin.
Conflict of Interest Disclosures: All authors have completed and submitted the ICMJE Form for Disclosure of Potential Conflicts of Interest and none were reported. Dr Lin reports being a consultant to Cochlear and receiving speaker honoraria from Amplifon.
Funding/Support: This study was funded in part by the Eleanor Schwartz Charitable Foundation and supported by grants NIH R33DC015062 (Dr Lin) and NIH KL2TR001077 (Dr Reed) from the National Institutes of Health.
Role of the Funder/Sponsor: The Eleanor Schwartz Charitable Foundation, National Institutes of Health, and Department of Health and Human Services had no role in the design and conduct of the study; collection, management, analysis, and interpretation of the data; preparation, review, or approval of the manuscript; and decision to submit the manuscript for publication.
Create a personal account or sign in to: