• We have developed a microsurgical spatula set consisting of four different tips and a universal handle. The set offers a range of tip angulation of 30°, 45°, 60°, and 80°. The bottom surfaces of the spatula tips are covered with fine diamond dust. The 30° tip has a blunt and tapered end. All tips can be introduced through a conventional 20-gauge sclerotomy in a pars plana vitrectomy. The tips fit on a universal handle. The set has been used as a complement to vitreous forceps in dissecting and removing preretinal membrane.
Tolentino FA, Wu G. Diamond-Dusted Microspatula Tips for Vitreoretinal Surgery. Arch Ophthalmol. 1987;105(12):1732–1733. doi:10.1001/archopht.1987.01060120130040
Coronavirus Resource Center
Customize your JAMA Network experience by selecting one or more topics from the list below.
Create a personal account or sign in to: