This article is only available in the PDF format. Download the PDF to view the article, as well as its associated figures and tables.
We read with interest the letter by Dr Fry. It is quite possible that a.12 forceps could be another source of scratch marks on an acrylic intraocular lens. In these cases, one would expect to see 1 line and not 2, as was described in our case.The modifications in Dr Fry's technique are important, namely, to enlarge the wound as well as to avoid contact between the intraocular lens implant and a device that could scratch it.
Vrabec MP, Syverud JC. Another Possible Cause of Forceps-Induced Scratching of a Foldable Acrylic Intraocular Lens-Reply. Arch Ophthalmol. 1997;115(6):823. doi:10.1001/archopht.1997.01100150825034
Coronavirus Resource Center
Customize your JAMA Network experience by selecting one or more topics from the list below.