This article is only available in the PDF format. Download the PDF to view the article, as well as its associated figures and tables.
The ligator described here is especially designed for tying bleeding points which are difficult to tie with the surgeon's fingers.
This instrument is composed of a main shaft into which a lower segment is welded. The segment is brought to a sloping point, in the end of which is a small opening slightly larger than a no. 2 catgut, through which the ligature is threaded. About 3 inches (7.6 cm.) from this tip there are another opening and a guide where the ligature makes its exit. On the under side of this portion of the shaft is a
A new instrument.
serrated flexible steel clip for attaching the ligator to the shank of a hemostat and receiving the loose ends of the ligature.
On the main shaft is a carriage, on which a clip is fastened to receive the ligature after it is brought through the exit. This
FILE NW. A NEW LIGATOR. Arch Otolaryngol. 1933;17(2):243–244. doi:10.1001/archotol.1933.03570050230010
Coronavirus Resource Center
Customize your JAMA Network experience by selecting one or more topics from the list below.
Create a personal account or sign in to: