An almost infinite variety of controlled destructive effects is made available to the surgeon by modulation of high frequency oscillating currents. The character of the effect produced is not determined by the frequency of the oscillations, but appears to depend entirely on the character of the oscillatory wave trains utilized. When a surgeon desires to destroy tissue en masse by heating, a train of damped oscillations is customarily used; but to obtain cutting effects, a train of undamped waves is used, although tissue can also be readily heated by undamped waves. A damped wave is one with an oscillating potential which tends to return to zero, as shown in figure 1 b; an undamped wave is one with an oscillating potential which remains constant, as shown in figure 1 a. The fundamental arrangements for production of high frequency damped oscillations (condenser-gap-inductance) and undamped oscillations (e. g., triode tube), which are
McLEAN AJ. THE BOVIE ELECTROSURGICAL CURRENT GENERATORSOME UNDERLYING PRINCIPLES AND RESULTS. Arch Surg. 1929;18(4):1863–1873. doi:10.1001/archsurg.1929.01140130965064
Customize your JAMA Network experience by selecting one or more topics from the list below.